RT-qPCR is the current recommended laboratory method to diagnose SARS-CoV-2 acute infection, several factors such as requirement of special equipment, time consuming, high cost and skilled staff limit the use of these techniques. A more rapid and high-throughput method is essential.
|Pir ELISA Kit| Rat Pirin ELISA Kit|
|REN ELISA Kit| Rat Renin ELISA Kit|
|Kl ELISA Kit| Rat Klotho ELISA Kit|
|LN ELISA Kit| Rat Laminin ELISA Kit|
|Lep ELISA Kit| Rat Leptin ELISA Kit|
We analyzed clinical data and nasopharyngeal samples, collected during September 2020, from patients attended at the emergency department of a secondary hospital and in two primary healthcare centers in Madrid. The performance of the Panbio COVID-19 AG Rapid Test Device for the detection of SARS-CoV-2 antigen was compared to RT-qPCR.
255 nasopharyngeal swabs, including 150 from the emergency department and 105 from primary helthcare centers, were tested. 184 patients were symptomatic (72.1 %). Amongst the 60 positive RT-qPCR samples, 40 were detected by the rapid antigen test, given an overall sensitivity of 73.3 %. All the samples detected positive with the rapid antigen test were also positive with RT-qPCR. The median cycle threshold was 23.28 (IQR 18.5–30.16). Patients with less than seven days onset of symptoms showed a higher viral load, and sensitivity for rapid antigen test (86.5 %), compared to those with more days (sensitivity of 53.8 %)(p < 0.004).
The rapid antigen test evaluated in this study showed a high sensitivity and specificity in samples obtained during the first week of symptoms and with high viral loads. This assay seems to be an effective strategy for controlling the COVID-19 pandemic for the rapid identification and isolation of SARS-CoV-2 infected patients.
Ensuring accurate diagnosis is essential to limit the spread of SARS-CoV-2 and for the clinical management of COVID-19. Although real-time reverse transcription polymerase chain reaction (RT-qPCR) is the currently recommended laboratory method to diagnose SARS-CoV-2 acute infection, several factors such as requirement of special equipment, time consuming, high cost and skilled staff limit the use of these molecular techniques. A more rapid and high-throughput method is in growing demand.
Until now, the use of antigen detection tests alone had been ruled out and not recommended due to their low sensitivity . Previously in the first wave, several easy to perform rapid antigen detection tests were developed as the first line of diagnostic. However, the results obtained were not good enough.
Background: Antigen-detecting rapid diagnostic tests (Ag-RDTs) for the detection of SARS-CoV-2 offer new opportunities for testing in the context of the COVID-19 pandemic. Nasopharyngeal swabs (NPS) are the reference sample type, but oropharyngeal swabs (OPS) may be a more acceptable sample type in some patients.
Methods: We conducted a prospective study in a single screening center to assess the diagnostic performance of the Panbio COVID-19 Ag Rapid Test (Abbott) on OPS compared with reverse-transcription quantitative PCR (RT-qPCR) using NPS during the second pandemic wave in Switzerland.
Results: 402 outpatients were enrolled in a COVID-19 screening center, of whom 168 (41.8%) had a positive RT-qPCR test. The oropharyngeal Ag-RDT clinical sensitivity compared to nasopharyngeal RT-qPCR was 81% (95%CI: 74.2-86.6). Two false positives were noted out of the 234 RT-qPCR negative individuals, which resulted in a clinical specificity of 99.1% (95%CI: 96.9-99.9) for the Ag-RDT. For cycle threshold values ≤ 26.7 (≥ 1E6 SARS-CoV-2 genomes copies/mL, a presumed cut-off for infectious virus), 96.3% sensitivity (95%CI: 90.7-99.0%) was obtained with the Ag-RDT using OPS.
Interpretation: Based on our findings, the diagnostic performance of the Panbio™ Covid-19 RDT with OPS samples, if taken by a trained person and high requirements regarding quality of the specimen, meet the criteria required by the WHO for Ag-RDTs (sensitivity ≥80% and specificity ≥97%) in a high incidence setting in symptomatic individuals.
|Paraffin Wax Dispenser|
|Paraffin wax, granular (56 - 60)|
|Paraffin wax, granular (56 - 60)|
|8KG Histoplast PE Paraffin|
|Paraffin oil, BP, Ph. Eur. grade|